Temperature_measurement

Temperature_measurement - Temperature Measurement Author...

This preview shows pages 1–2. Sign up to view the full content.

Temperature Measurement Author: John M. Cimbala, Penn State University Latest revision: 14 November 2007 [minor modification 09 April 2008] Introduction There are four basic types of temperature measuring devices, each of which uses a different principle: o Mechanical (liquid-in-glass thermometers, bimetallic strips, etc.). o Thermojunctive (thermocouples). o Thermoresistive (RTDs and thermistors). o Radiative (infrared and optical pyrometers). Each of these is defined and discussed in this learning module, with most of the emphasis placed on thermojunctive temperature measuring devices – thermocouples. Mechanical temperature measuring devices Principle of operation : o A change in temperature causes some kind of mechanical motion, typically due to the fact that most materials expand with a rise in temperature . Mechanical thermometers can be constructed that use liquids, solids, or even gases as the temperature-sensitive material. o The mechanical motion is read on a physical scale to infer the temperature. Liquid-in-glass thermometer o The most common and well-known thermometer is the liquid-in-glass thermometer . o As the temperature rises, the liquid expands, moving up the tube. The scale is calibrated to read temperature directly. Scale Bulb Metal A Metal B o Usually, mercury or some kind of alcohol is used for the liquid. Bimetallic strip thermometer o Two dissimilar metals are bonded together into what is called a bimetallic strip , as sketched to the right. o Suppose metal A has a smaller coefficient of thermal expansion than does metal B. o As temperature increases, metal B expands more than does metal A, causing the bimetallic strip to curl upwards as sketched. o One common application of bimetallic strips is in home thermostats, where a bimetallic strip is used as the arm of a switch between electrical contacts. As the room temperature changes, the bimetallic strip bends as discussed above. When the bimetallic strip bends far enough, it makes contact with electrical leads that turn the heat or air conditioning on or off. Metal A Metal B o Another common application is for use as oven thermometers or wood burner thermometers. These thermometers consist of a bimetallic strip wound up in a spiral , attached to a dial that is calibrated into a temperature scale. Pressure thermometer o A pressure thermometer , while still considered mechanical, operates by the expansion of a gas instead of a liquid or s ( Note : There are also pressure thermometers that use a liquid instead of a gas.) olid. o Suppose the gas inside the bulb and tube can be considered an ideal gas. The ideal gas law is P V = mRT , where P is the pressure, V is the volume of the gas, m is the mass of the gas, R is the gas constant for the specific gas (not the universal gas constant), and T is the absolute temperature of the gas.

This preview has intentionally blurred sections. Sign up to view the full version.

View Full Document
This is the end of the preview. Sign up to access the rest of the document.

{[ snackBarMessage ]}

Page1 / 8

Temperature_measurement - Temperature Measurement Author...

This preview shows document pages 1 - 2. Sign up to view the full document.

View Full Document
Ask a homework question - tutors are online