{[ promptMessage ]}

Bookmark it

{[ promptMessage ]}

2LB_Lab_7_DCElecCircuits11

# 2LB_Lab_7_DCElecCircuits11 - Physics 2LB DC Electrical...

This preview shows pages 1–2. Sign up to view the full content.

Physics 2LB DC Electrical Circuits (*** Includes Pre Lab Assignment) Materials: Two digital multimeters, 6 V battery, resistor-diode circuit board, potentiometer, Computer with Science Workshop 750 Interface and hook-up cables. Purpose: To become familiar with digital multimeters used for measurement of voltage, current, and resistance; to test Ohm's Law; to measure the I-V characteristics of non-linear circuit elements. INTRODUCTION An ideal device for measurement of voltage, current, or their ratio (resistance) in a circuit, would do so without in any way perturbing the circuit being measured. Real devices can approach this ideal in varying degrees. Thus a reasonably rugged and portable old style voltmeter, based upon a moving coil galvanometer, would typically draw one milliampere of current at full-scale deflection. If a range of 0–1 V was to be covered, a series resistance of 1 k (including coil resistance) would be connected to the coil to complete the device. The input impedance of this voltmeter would then be 1 k , low enough to lead to serious error in the measurement of voltage sources having output impedances of this order or higher. A modern voltmeter (Digital Multimeter) based upon solid-state circuitry will have an input impedance of 10 7 - 10 12 . The digital multimeter to be used here has high enough input impedance that for purposes of voltage measurement in this experiment , it may be considered as ideal. Current measurement can be accomplished by inserting a moving-coil galvanometer into the circuit. The galvanometer will include low value shunt resistors for higher current ranges. An alternative would be to measure the voltage drop across a low value resistor using a high impedance solid-state device. In either case an appreciable series resistance must be introduced into the circuit in which current is being measured. This resistance will be evaluated in a typical situation below. Resistance could be measured by a bridge technique or, as will be the case here, by passing a known current through the unknown resistor and measuring the voltage drop across it. Accuracy of this method requires both a high impedance current source and a high impedance voltmeter. For purposes of this experiment both conditions are satisfactorily met by the digital multimeters to be used. PROCEDURE A. Basic Measurements of Voltage, Resistance, and Current Using a Digital Multimeter (DMM) [1.5] 1. Set the DMM for DC voltage measurements by rotating the function switch until the black dot on the rotating bar is positioned at the 20 DCV range. Connect the positive (+) terminal of a 6 V battery to the red (V- -Hz) input connector and the negative ( - ) terminal to the black (COM) connector of the DMM.

This preview has intentionally blurred sections. Sign up to view the full version.

View Full Document
This is the end of the preview. Sign up to access the rest of the document.

{[ snackBarMessage ]}