This preview shows page 1. Sign up to view the full content.
Unformatted text preview: the TC Cold Junction Source to "Constant" and the Cold Junction Temperature (Degrees) to the lab room temperature as you measure it (20-25 C). Use the temperature reading from the thermometer placed on top of the ADC input box. In the main DMM window set the Range to Auto, Mode to Volts DC and when ready click the Run Continuously button as shown in Figure 4. You may also switch to a temperature reading mode by clicking on the thermometer icon (see Figure 5). Figure 2. The DMM General Settings Window. Note: your computer may contain a different Device than the one shown here. 16 Page 3 of 16...
View Full Document
- Spring '09