This preview shows page 1. Sign up to view the full content.
Unformatted text preview: Modify the VI such that the user specifies a sampling frequency instead.) d. Display the acquired voltage on a chart as well as on a digital indicator. e. Track errors and indicate if an error occurs. Also, modify your VI as specified below a. Set the input voltage range (maximum and minimum voltages) to +/- 10 V. b. Change the Waveform Chart label to Analog Input Chart. c. Change the digital indicator label to Analog Input Value, set to 3 Digits of Precision. d. Save VI as yourname_lab2_1Channel.vi (rev. 2006)...
View Full Document
- Spring '08