introduction

# 003 k thus the value to be recorded is 673 3 the

This preview shows page 1. Sign up to view the full content.

This is the end of the preview. Sign up to access the rest of the document.

Unformatted text preview: is uncertain by 0.673 × 0.002 (0.2%) + 0.002 (2 digits) ≈ 0.001 + 0.002 or ±0.003 kΩ. Thus the value to be recorded is 673 ± 3 Ω. The uncertainties for the Fluke meters are given in the following tables. Fig (4) shows the errors that should be applied to dc volts and resistance readings, Fig (5) should be used for ac voltages and Fig (6) is used for both ac and dc current measurements. 4 80 Series III Users Manual Table 13. DC Voltage, Resistance, and Conductance Function Specifications Accuracy1 Function Range Resolution Model 83 Model 85 Model 87 4.000 V 40.00 V 400.0 V 1000 V ±(0.1% + 1) ±(0.1% + 1) ±(0.1% + 1) ±(0.1% + 1) ±(0.08% + 1) ±(0.08% + 1) ±(0.08% + 1) ±(0.08% + 1) ±(0.05% + 1) ±(0.05% + 1) ±(0.05% + 1) ±(0.05% + 1) 0.1 mV ±(0.3% + 1) ±(0.1% + 1) ±(0.1% + 1) 400.0 Ω 4.000 kΩ 40.00 kΩ 400.0 kΩ 4.000 MΩ 40.00 MΩ 40.00 nS F 0.001 V 0.01 V 0.1 V 1V 400.0 mV L 0.1 Ω 0.001 kΩ 0.01 kΩ 0.1 kΩ 0.001 MΩ 0.01 MΩ 0.01 nS ±(0.4% + 2)2...
View Full Document

Ask a homework question - tutors are online