This preview shows pages 1–2. Sign up to view the full content.
This preview has intentionally blurred sections. Sign up to view the full version.View Full Document
Unformatted text preview: 16) Another beep signifies the end of the test. 17) Load is automatically lifted. 18) Screens shows 8 data test points as pre-set and more detailed results are revealed using the green folder icon (top right corner of the screen). 19) Print the results. 20) Clean the piston, and barrel using cloth squares. 21) Clean the die using drill bit provided. 22) Plunge the barrel using cut cloth and plunger while the barrel is still warm. 23) Replace the clean die into the barrel. 24) Push the encoder down into secure position. 25) Replace all materials and return the lab equipment to original place. NOTE : If Modular unit does not turn on check the circuit breaker that is in the corner to the right of the Instron to see if it is tripped. If it is, reset the top breaker in the box. If the above does not work take the casing off of the rear of the unit and check the fuses. If fuse needs replaced, replace it with a 10 amp fuse....
View Full Document
This note was uploaded on 04/09/2008 for the course PET 375 taught by Professor Ibeh during the Spring '08 term at Pittsburg State Uiversity.
- Spring '08