100%(4)4 out of 4 people found this document helpful
This preview shows 10 out of 12 pages.
4EXPERIMENT 1Flow Through a Venturi MeterINTRODUCTIONThe aim of this experiment is to provide a better understanding of how to apply Bernoulli'sequation in real flows, and of the physical meaning of the terms static pressure, stagnation pressure anddynamic pressure.The venturi meter is a flow-measurement device consisting of a convergent-divergent length ofpipe. The experimental setup consists of a bell-mouth inlet through which air is drawn from the room,followed by the venturi. A vacuum cleaner connected to the downstream end of the venturi providessuction. The bell-mouth inlet is itself a flow-measurement device and in this experiment the flow ratesdetermined from it will be assumed to be the true values. The venturi is instrumented with static pressuretaps at a number of locations.Because of the change in area, the velocity, and therefore the static pressure, changes along theventuri. Assuming one-dimensional flow, conservation of mass and Bernoulli's equation can be used tocalculate the pressure variation along the venturi. The flow rate can also be expressed as a function ofthe pressure difference between the venturi inlet and the throat; this will be referred to as the "ideal flowrate" for the venturi. Recall from lectures that Bernoulli’s equation is an energy equation which can onlybe applied along a streamline when there is no friction (ie. no viscous effects). Because of the no-slipcondition, the flow near the walls of the venturi is viscous. This effect spreads further and further into theflow with downstream distance. Bernoulli’s equation will only be valid in the core of the flow where viscouseffects have not reached. Nor will the flow be one-dimensional. Therefore, the simple analysis describedearlier will only approximate the real flow behaviour. One of the goals of the experiment is to examinehow much the real flow differs from this approximate analysis.